Bigonoff () La Programmation des PIC par Bigonoff. Seconde Partie ( PIC16F 16F), 7ème Révision. I allready red the tutorial of “bigonoff”, where I found the explanation, but I’m using an 16F or an 16FA . does the code change??). Context: PIC 16F, MPLAB , ICD2 USB, Windows communicating with a debug executive ().
|Published (Last):||14 April 2015|
|PDF File Size:||2.30 Mb|
|ePub File Size:||11.72 Mb|
|Price:||Free* [*Free Regsitration Required]|
Is there some hints to connect easily and in a reproductive manner? It is possible to still use HS at a lower frequency.
Forum Themes Elegant Mobile. No need to be fancy, just an overview.
ICD2 – ICD0154
Back on the air. Any advices on the initialization sequence?
A PiC with floating inputs could drain too much current if inputs float to voltages neither high nor low. Bigonoff pic 16f pdf Essentials Only Full Version.
You don’t ever state that you’ve tried other devices.
I’ll keep you informed. Thanks anyway Microchip for this helpful forum. Gixy Super Member Total Posts: High frequency current changes just affect the PS in such a way that a lot of noise is induced into the power supply lines, so all devices run from a noisy power source.
Impossible to determine a reproductible sequence Microchip’s PIC microcontrollers with intelligent analog integration like the PIC16FX family make it possible to get advanced features such as digital control to smooth dimming, as well as color temperature tuning, usagebased lifetime prediction and networked communication for. Absolutly right Vito, but For example, the state they power-up. The ICD2 is somewhat sensitive to noisy environments.
Hi, HS just provides a higher gain of the internal inverter of the oscillator circuit.
sample code RS232
How to program a PIC microcontroller. For more details and other Windows versions.
What do you think about the init sequence? Have you switched devices? H H Administrator Total Posts: It works without any problem. The cable between target and ICD2 is 20cm long. The final board is finished and still the same problem: Hi all, maybe a hint: Hi, Are you sure everything is working correctly at 20MHz?
ICD2 – ICD | Microchip
Debug breakpoints automatically disabled Future of PIC32 bigoonff cannot set internal oscillator frequency beyond It was a slight code modification to the firmware on the ICD2. Furthermore, it is not so easy to determine the exact sequence to make ICD2 init correctly. Imasson alias Laurent wrote me that he tried many changes on his board, and he never exceed 16 MHz for debugging The target is supplied by a lab PS and I’m sure also that there is a big cap in it.
There is now a lot of decoupling caps, oscillator works scopebut always the same message The Adode and Acrobat trademarks and copyrights are used for comparison and reference for the users only; they belong to Adobe Systems Inc and can be found at the following url:.
No matter the state, but avoid input pins to float.